BEVISTON CMS disposes of a stainless steel bottom including a rubber edge providing stability and slip resistance when working with it.
For correct dosage fill either the products Bevisto W1 (5 %) or Bevisto W2 (2 %) from the respective unit (e. g. dosage bottle or can) into BEVISTON CMS and fill up with 2 litres of warm
water (temperature: about 40 – 60 °C). Read off the filling level from the scale of BEVISTON CMS.
Afterwards connect the aspiration tubes to the adapters of BEVISTON CMS mentioned above so that the respective cleaning solution (either Bevisto W1 or Bevisto W2) can be rinsed into the
After 1 litre of cleaning liquid has been rinsed through the aspiration tubes into the aspiration system, fill the remaining litre contained in BEVISTON CMS carefully and gradually into
the spittoon bowl. This is for maintaining the spittoon bowl and the spittoon draining filter unit underneath as well as the further aspiration tubes.
We recommend carrying out the cleaning of the dental aspiration lines at least twice a day (at noon and in the evening). When starting work the next day we recommend rinsing again the
aspiration lines with water (e. g. by operating the spittoon bowl rinsing). As a result residues detached by Bevisto W1 and or Bevisto W2 overnight will be rinsed out.
We recommend checking, cleaning or exchanging the filters of the appropriate dental unit and/ or the dental aspiration system daily for detached dirt residues (like e. g. detached
biofilm residues, etc.) in accordance to the instructions of the respective manufacturer.
At the outer wall of the container there is a measuring scale as well as icons describing the handling of the container in pictures and words.